AGILENT N5700 DRIVER DOWNLOAD
CLEar which shows the use of the semicolon separating the two commands, and also illustrates the command path concept. This register is a mask for enabling specific bits from the Questionable Event register to set the questionable summary bit QUES of the Status Byte register. This means that you can remotely program the Agilent N power supplies using the same commands that are used to program the xA power supplies. Language Reference 5 System Commands System commands control system functions that are not directly related to output control, measurement, or status functions. Error List The following table documents the various error messages that the power supply supports: Is the AC input voltage within range? Let the oscilloscope run for a few seconds to generate enough measurement points.
|Date Added:||16 November 2011|
|File Size:||33.82 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Press and hold the LAN button for three seconds. Programming Examples 6 ‘ Set the over voltage level. An example of a message with two commands is: The immediate level is the output current setting. On the Agilent Infiniium scope, the maximum peak-to-peak voltage measurement is indicated at the bottom of the screen on aiglent right side. Configure the following LAN address parameters.
The values are programmed in amperes.
N5700 and N8700 Series DC System Power Supplies, GPIB, Single Output
All settings NOTE commands have a corresponding query. Let the oscilloscope run for a few seconds to generate enough measurement points. Page 23 Installation 2 Install the shield after you have finished connecting the load wires.
A check-out procedure is included to let you verify that the power supply is operating properly. LEVel] Calibrates the output current programming: It returns a string of four fields separated by commas. Appendix D Compatibility Differences — In General The following table documents the general differences between the way Compatibility commands work on the Agilent N power supplies and the way they worked on the Agilent xA power supplies.
The programming range and monitoring signal range can be selected using the SW1 setup switch. AC peaks of the output current ripple, several DC measurements should be made and averaged.
Check AC input voltage. Appendix A Specifications Outline Diagram In most cases you can connect your power supply to any one of these interfaces and be up and running with a minimum amount of configuration. Safety Notices Terminal is at earth impaired if it is used in a manner not product. Attach a tag to the unit identifying the owner and indicating the required service or repair. This means that you can remotely program the Agilent N power supplies using the same commands that are used to program the xA power supplies.
Output Programming Example 6 Programming Examples Output Programming Example This program sets the voltage, current, over-voltage, and the over- current protection.
This could damage the power supply. Verification These procedures calibrate the power supply.
Keysight Technologies (Agilent HP) N Series – W used or new for sale at used-line
All pins are optically isolated from the power supply output. The display first scrolls through the four segments of the IP address, followed by the six segments of the Ethernet EA address.
The VISA address is: Parallel Connections 2 Installation Battery Charging If a battery or external voltage source is connected across the output and the CAUTION output is programmed below the battery or external voltage source, the power supply will continuously sink current from the external source.
All compatibility commands are accepted; however, some commands do nothing. This DoC applies to above-listed products placed on the EU market after: The second parameter is the password.
Adjust the resistors to change the power supply output. A password is required if calibration mode is being enabled and the existing password is not 0. Actual repair is done through unit exchange. Connect each load to the distribution terminals separately. Don’t show me this message again.